Purchase this article with an account.
Xiao Wei Tan, A. Promoda P. Perera, Anna Tan, Donald Tan, K. A. Khor, Roger W. Beuerman, Jodhbir S. Mehta; Comparison of Candidate Materials for a Synthetic Osteo-Odonto Keratoprosthesis Device. Invest. Ophthalmol. Vis. Sci. 2011;52(1):21-29. doi: https://doi.org/10.1167/iovs.10-6186.
Download citation file:
© ARVO (1962-2015); The Authors (2016-present)
Osteo-odonto keratoprosthesis is one of the most successful forms of keratoprosthesis surgery for end-stage corneal and ocular surface disease. There is a lack of detailed comparison studies on the biocompatibilities of different materials used in keratoprosthesis. The aim of this investigation was to compare synthetic bioinert materials used for keratoprosthesis surgery with hydroxyapatite (HA) as a reference.
Test materials were sintered titanium oxide (TiO2), aluminum oxide (Al2O3), and yttria-stabilized zirconia (YSZ) with density >95%. Bacterial adhesion on the substrates was evaluated using scanning electron microscopy and the spread plate method. Surface properties of the implant discs were scanned using optical microscopy. Human keratocyte attachment and proliferation rates were assessed by cell counting and MTT assay at different time points. Morphologic analysis and immunoblotting were used to evaluate focal adhesion formation, whereas cell adhesion force was measured with a multimode atomic force microscope.
The authors found that bacterial adhesion on the TiO2, Al2O3, and YSZ surfaces were lower than that on HA substrates. TiO2 significantly promoted keratocyte proliferation and viability compared with HA, Al2O3, and YSZ. Immunofluorescent imaging analyses, immunoblotting, and atomic force microscope measurement revealed that TiO2 surfaces enhanced cell spreading and cell adhesion compared with HA and Al2O3.
TiO2 is the most suitable replacement candidate for use as skirt material because it enhanced cell functions and reduced bacterial adhesion. This would, in turn, enhance tissue integration and reduce device failure rates during keratoprosthesis surgery.
This PDF is available to Subscribers Only