Purchase this article with an account.
Qianying Gao, Xiang Chen, Jian Ge, Yongji Liu, Zhaoxin Jiang, Zhi Lin, Yaqin Liu; Refractive Shifts in Four Selected Artificial Vitreous Substitutes Based on Gullstrand-Emsley and Liou-Brennan Schematic Eyes. Invest. Ophthalmol. Vis. Sci. 2009;50(7):3529-3534. doi: https://doi.org/10.1167/iovs.08-2802.
Download citation file:
© ARVO (1962-2015); The Authors (2016-present)
purpose. To determine and compare the refractive shifts based on Gullstrand-Emsley and Liou-Brennan schematic eyes after filling them with four selected artificial vitreous substitutes: silicone oil, heavy silicone oil, hydrogels, and encapsuled balanced salt solution.
methods. The optical constants of artificial vitreous body–filled eyes were calculated based on Gullstrand-Emsley and Liou-Brennan schematic eyes with accommodation relaxed. The theoretical refractive shifts in these two models were compared in pars plana vitrectomy (PPV), PPV plus lensectomized and PPV plus intraocular lens (IOL) eyes after four artificial vitreous tamponades.
results. The Gullstrand-Emsley schematic eye shows refractive shifts of +8.710, −4.544, +1.136, and −0.338 D in PPV eyes; +11.044, +20.332, +16.351, and +17.413 D in PPV plus lensectomized eyes; and the need for IOL powers of +22.195, +22.366, +22.292, and +22.312 D in PPV plus IOL eyes in silicone oil, heavy silicone oil, hydrogels, and encapsuled balanced salt solution tamponade eyes, respectively. Similarly, the Liou-Brennan schematic eye induced shifts of +6.260, −3.266, +0.817, and −0.272 D in PPV eyes; +13.181, +20.654, +17.451, and +18.305 D in PPV plus lensectomized eyes; and the need IOL powers of +13.522, +23.767, +19.389, and +20.558 D in PPV plus IOL eyes, respectively.
conclusions. The Gullstrand-Emsley schematic eye is a convenient and accurate model for predicting refractive shifts for hydrogels and encapsuled balanced salt solution substitutes in PPV eyes. The Liou-Brennan schematic eye is recommended for silicone oil and heavy silicone oil in PPV eyes and for all four substitutes in PPV plus lensectomized eyes and PPV plus IOL eyes. In addition, the encapsuled balanced salt solution changes the refraction little in either schematic eye.
This PDF is available to Subscribers Only