Purchase this article with an account.
Juliane Matlach, Martin Wagner, Uwe Malzahn, Winfried Göbel; Repeatability of Peripapillary Retinal Nerve Fiber Layer and Inner Retinal Thickness Among Two Spectral Domain Optical Coherence Tomography Devices. Invest. Ophthalmol. Vis. Sci. 2014;55(10):6536-6546. doi: 10.1167/iovs.14-15072.
Download citation file:
© ARVO (1962-2015); The Authors (2016-present)
To compare measurement of macular inner retina and peripapillary retinal nerve fiber layer (pRNFL) thickness using two spectral-domain optical coherence tomography (SD-OCT) devices in glaucoma patients, patients with ocular hypertension, idiopathic and atypical Parkinson disease, and healthy controls.
A total of 171 eyes of 146 participants underwent successful pRNFL and macular scanning and automated measurement of ganglion cell layer + inner plexiform layer (GCL-IPL) using Cirrus HD-OCT or retinal nerve fiber layer + GCL-IPL (RNFL–GCL-IPL) using RTVue-100. Macular RNFL was added to the GCL-IPL thickness measured by Cirrus and was compared to the RNFL–GCL-IPL thickness of the RTVue in the corresponding Cirrus sectors. Intraclass correlation coefficient (ICC) was calculated to determine repeatability of three consecutive measurements; ICC and Bland-Altman analysis to assess agreement between OCTs; Pearson's correlation coefficient to assess strength of linear correlation.
Repeatability of average macular RNFL–GCL-IPL thickness measurement was excellent with an ICC of 0.998 for Cirrus and 0.995 for RTVue. Repeatability was also good for average pRNFL thickness measurements. Both instruments demonstrated a good consistency in measurements with ICC values ranging from 0.844 to 0.862 for macular RNFL–GCL-IPL and 0.718 to 0.958 for pRNFL thickness.
Measurement of pRNFL and macular RNFL–GCL-IPL thickness has a high degree of repeatability for both OCT devices. Despite a high correlation between measurements of the two OCT devices and fair to excellent ICC values representing a high consistency in the measurements of the two devices, RTVue measured a thicker macular RNFL–GCL-IPL and pRNFL thickness.
This PDF is available to Subscribers Only