Purchase this article with an account.
K. Rohrer, B. E. Frueh, I. Clemetson, R. Waelti, D. Goldblum; New Non-Contact Biometer. Invest. Ophthalmol. Vis. Sci. 2008;49(13):388. doi: https://doi.org/.
Download citation file:
© ARVO (1962-2015); The Authors (2016-present)
To evaluate and compare axial length (AL), anterior chamber depth (ACD), lens thickness (LT), and central corneal thickness (CCT) measurements assessed by optical biometry (OB), ultrasound biometry (UB), optical low coherence reflectometry pachymetry (OLCR-P), and by a new prototype applying optical low coherence reflectometry biometry (OLCR-B).
OLCR-B (Haag-Streit), OLCR-P (Haag-Streit), OB (Zeiss), and UB (Tomey) were performed in 35 eyes of 20 patients with incipient cataract. Paired t-test was performed.
The mean AL was for OLCR-B 24.35 mm ± 2.32 (SD), OB 24.33 mm ± 2.41, and UB 24.03 mm ± 2.27; the mean ACD for OLCR-B 3.10 mm ± 0.51, OB 3.15 mm ± 0.52, and UB 3.09 mm ± 0.47; the mean LT for OLCR-B 4.71 mm ± 0.42, and UB 4.73 mm ± 0.49; the mean CCT for OLCR-B 543 µm ± 44, OLCR-P 536 µm ± 32, and UB 541 µm ± 41. No statistical significant differences were found between the groups, except in the mean AL comparing OLCR-B vs. UB; and mean AL comparing OB vs. UB (p < 0.05) .
OLCR-B, OLCR-P, OB and UB give comparable results in patients with incipient cataract. There are differences in the eye length measurements comparing OLCR-B and OB with UB. The advantages of the OLCR-B prototype are the non contact method, the exact measurement of AL, ACD, LT and CCT on the same interferometry method and hence the short duration of the measurements.
Clinical Trial: :
This PDF is available to Subscribers Only