Purchase this article with an account.
Y. Li, O. Tan, M. Tang, D. Huang; Corneal Epithelial Thickness Mapping With Fourier-Domain Optical Coherence Tomography. Invest. Ophthalmol. Vis. Sci. 2008;49(13):2813. doi: https://doi.org/.
Download citation file:
© ARVO (1962-2015); The Authors (2016-present)
To map corneal epithelial thickness with optical coherence tomography (OCT).
A Fourier-domain OCT system (RTVue, Optovue Inc., Fremont, CA) with a corneal adaptor module (CAM) was used to image the cornea. It operated at 0.8 µm wavelength and had a scan rate of 26,000 axial-scan/second. The axial resolution was 5 µm in tissue. A pachymetry scan pattern (8 radials, 1024 axial scans each, 6mm diameter, repeat 3 times) centered at the corneal vertex was used to map the thickness of the cornea and its sub-layers. The three repeated scans in a scan set were registered and averaged before epithelial boundary detection. The epithelial thickness (tear film included) map was divided into 3 zones by diameter: central 2 mm, superior 2-5 mm, and inferior 2-5 mm. The average epithelial thickness value from each zone was calculated. 24 eyes of 12 normal subjects were scanned 4 times. The repeatability of the measurement was evaluated by the pooled standard deviation (SD).
The central, superior, and inferior epithelial thickness averages (± population SD) were 53.5 ± 2.4 µm, 49.4 ± 2.4 µm, 50.1 ± 2.5 µm. There was no significant differences between right and left eyes (T-test, p=0.69, 0.37, 0.31). The epithelium was thicker in the center than in the superior (mean difference 3.8 ± 1.3 µm, p<0.001) and inferior zones (3.2 ± 1.1 µm, p<0.001). The inferior epithelium was slightly thicker than the superior epithelium (mean difference 0.6 ± 1 µm, p<0.001). The repeatability of the measurements was 0.7 µm in all three zones.
High-resolution high-speed FD-OCT is able to map the corneal epithelial thickness with excellent reproducibility. The averaged epithelial thickness pattern is consistent with previous literature based on ultrahigh frequency ultrasound.
This PDF is available to Subscribers Only